Sample Rate: 100 MS/s per channel. Effective sample rates up to 2 GS/s in repetitive storage mode (0.5 µs/div and faster in single-channel mode, 0.2 µs/div and faster dual-channel).
Resolution: Vertical: 8-Bits (25 levels per division), up to 12-Bits in average mode. Horizontal: 10-Bits (100 points per division), 9-Bits per channel in dual channel mode.
Record Length: 4K or 1K selectable. 2K or 512 per channel in dual channel mode.
Pre/Post Trigger: 1/8, 1/2, or 7/8 trigger position selectable, point-selectable via menu.
Acquisition Modes: Peak Detect (10 ns glitch capture at all available sweep speeds); Accumulated Peak Detect; Average (weight-selectable from 1/1 to 1/256 in a binary sequence); and Sample.
Save Reference Memory: One 4K or three 1K acquisitions. Battery-backed memory stores waveforms for up to 3 years.
Additional: 26K of extended memory (store up to 26 waveform sets).
Bandwidth (-3 dB) and Rise Time: 100 MHz and 3.5 ns (0 to +35 degrees C); 80 MHz and 4.4 ns (2 mV/div or +35 to +50 degrees C).
Deflection Factor and Accuracy: 2 mV/div to 5 V/div ± 2% (+15 to +35 degrees C); ± 3% (0 to +50 degrees C).
Vertical Operating Modes: CH 1, CH 2, CH 2 INVERT, ADD, ALT, CHOP (500 kHz), and XY.
CMRR: At least 10:1 at 50 MHz.
Input R and C: 1 Megohm 20 pF.
Max Input Voltage: 400 V (DC + peak AC), 800 V p-p.
Channel Isolation: 100:1 at 50 MHz.
Sweep Speeds: A sweep: 0.5 s/div to 0.05 µs/div, extended to 5 ns/div with X10 magnification. 5 s/div to 0.05 µs/div in store mode.
Additional: B Sweep: 50 µs/div to 0.05 µs/div.
Accuracy: Nonstore Mode: X1: ± 2%; X10: ± 3% (+15 to +35 degrees C). X1: ± 3%; X10: ± 4% (0 to +50 degrees C). Store Mode: ± 0.1% over full 10.24 divisions.
Horizontal Operating Modes: Nonstore Mode: A, ALT (A intensified by B and B), B. Store Mode: A, A intensified by B, B, 4K COMPRESS.
Delay Jitter: 5000:1.
Delay Time Accuracy: ± 1% (+15 to +35 degrees C); ± 2% (0 to +50 degrees C).
Trigger Sensitivity (A and B):
Internal: 0.35 div at 10 MHz, 1.5 div at 100 MHz.
External: 40 mV at 10 MHz, 150 mV at 100 MHz.
Trigger Operating Modes: A-Mode: P-P AUTO (also for TV LINE), NORM, TV FIELD, SGL SWP.
Additional: B-Mode: Runs-After-Delay, Triggered-After-Delay.
Trigger Source: A Trigger: VERT MODE, CH 1, CH 2, LINE, EXT.
Additional: B Trigger: VERT MODE, CH 1, CH 2.
With Internal Source: AC with P-P AUTO, TV LINE, or TV FIELD mode; DC with NORM or SGL SWP mode.
With external source: AC, DC, or DC/10. With either source: HF REJECT (attenuates above 40 kHz), LF REJECT (attenuates below 40 kHz).
Variable Holdoff: At least 10:1.
Deflection Factors: Same as vertical system.
X-Axis: 2.5 MHz in nonstore mode, up to 100 MHz in store mode.
Y-Axis: same as vertical system.
Phase Difference: ± 3 degrees from DC to 150 kHz.
Cursor Function and Accuracy: DeltaVolts: ± 3% of reading. DeltaTime: ± 1 display interval (5 s/div to 1 µs/div); ± [2 display intervals +500 ps] (0.5 µs/div to 0.05 µs/div).
X-Y Plotter Output: Plots all displayed waveforms, CRT readout, and graticule (selectable).
External Clock Input: DC to 1 kHz (roll mode), DC to 100 kHz (record mode).
ANSI/IEEE-488 GPIB Interface (Opt. 10): Function Subsets Implemented: SH1, AH1, T5, L3, SR1, RL2, PP0, DC1, DT0, C0, E2. Plotter Devices: HPGL, Epson FX-Series, HP ThinkJet. Data Transfer Rate: approximately 1 KB.
EIA Std RS-232C Interface (Opt. 12): Baud Rate: 50 to 2400 for interactive use, up to 4800 for driving plotters. Plotter Devices: HPGL, Epson FX-Series, HP ThinkJet. Connectors: DCE (female), DTE (male).
Display: 8 cm x 10 cm, 14 kV nominal voltage.
Controls: A INTENSITY, TRACE ROTATION, BEAM FIND, FOCUS, STORAGE/READOUT INTENSITY, GRATICULE ILLUMINATION.
Additional: B INTENSITY.
Z-Axis: 5 V causes noticeable modulation. Usable to 20 MHz.
咨询电话: 132- 4173- 5460 , 邮箱: 52634729@QQ.com